Ultra pulse CO2 laser (beauty) treatment machine is a CO2 laser treatment machine that can output ultra pulse laser. Its working principle, basic structure and performance are similar to ordinary CO2 laser treatment machine.
In order to facilitate clinical use, in addition to outputting ultra pulse laser, the ultra pulse CO2 laser (beauty) treatment machine is also compatible with all functions of ordinary CO2 laser treatment machine. It is a multifunctional CO2 laser treatment equipment.
The ultra pulse CO2 laser (beauty) treatment machine also has the following characteristics:
It can output pulsed laser with high energy and high repetition rate, meet the operation requirements of “laser selective photothermal effect”, quickly and effectively remove the diseased target tissue, minimize the damage of laser to normal tissue, and greatly improve the accuracy and safety of medical cosmetic clinical treatment.
Clinical practice has proved that in the same operation, the laser power used by pulse laser is much lower than that of continuous laser. Therefore, the tissue reaction caused by laser operation is light, the damage to the surrounding tissue is small, the cure time is shortened, the smoke produced during treatment is less, the field of vision is clear, and the harm to clinicians is reduced.
In order to obtain as fine a focusing spot as possible, the laser beam mode of the super pulse CO2 laser is usually better than that of the ordinary CO2 laser. The most ideal is to choose the CO2 laser with basic transverse model.
In order to obtain a better CO2 laser output mode, the inner diameter of the discharge tube of CO2 laser is required to be small, and the inner wall has strict concentricity and perpendicularity.
In order to ensure the need of high-energy and high pulse laser output, CO2 laser is required to have laser fast recovery characteristics in the structural design. For example, the laser resonator components should be made of vacuum materials with better resistance to electric shock and not easy to evaporate. At the same time, the quantity and proportion of laser gas and auxiliary gas should be strictly controlled in the process.
The shape and structure of the laser treatment machine should have a more humanized design, more coordinated with the surrounding environment, more convenient and safer to use.
Ultra pulse CO2 laser (beauty) treatment machine has the common characteristics of CO2 laser scalpel:
Longer output wavelength (10.6) μ m) , the photoelectric conversion efficiency is high, the output power is large, and after focusing, it can form a large power density, showing a sharp “knife edge”.
CO2 laser has the advantages of simple structure, mature technology, low cost, reliable performance and convenient operation and maintenance, which is conducive to popularization and application.
Compared with other laser scalpels, CO2 laser scalpel has the advantages of strong cutting force, large tissue absorption coefficient and small tissue penetration depth (about 0.23mm). It is not easy to hurt arteries and blood vessels during operation, but also has the functions of hemostasis and sealing blood vessels. It is especially suitable for fine surgery of medical cosmetology. It also has the advantages of clear operation field and not easy cross infection.